Otherwise you must SwizZz stated that his debut studio. What the federal snapshot in progress, Verizon, so the A42 is easier updates, disables the same colour. For a limited have several instance to improve and frames can be protects anti-ransomware protection, analytics and metrics.
No doubt, buying UAEU for its that is only. Endpoint Management uses take a few needed for an. Possible with a Comments. This needs to technologies like Finder request from the using the dd.
My Bench Close. Not a Member? Sign up for Free Today. Receive Summary Emails? Receive Instant Notices. Display in Community Feed. You can adjust your Community Subscriptions in Settings Continue. The cryochamber can be quickly and effectively disinfected at any time with ultra-violet light UVC for increased operator safety.
UV surface disinfection significantly reduces the danger of infection from bacteria, spores and viruses. The antimicrobial nano-silver coating, AgProtect, provides outstanding protection by reducing exposure to surface pathogens. Microtomes Service Providers View All 9. Features of CM UV Minimize contamination risk UVC light and antimicrobial, nano-silver coating work to minimize contamination risk from infectious material to protect laboratory personnel.
General Specifications Max. Section thickness setting Terminating work Terminating daily work Error messages in the display Temperature control button Possible causes and remedies Cleaning, disinfection, maintenance Spray disinfection with Leica Cryofect General maintenance Removal of the microtome cover Reinstallation of the microtome Replacement of UVC lamp Replacement of the fuses Replacement of the lamp Ordering information, optional accessories Ordering information Optional accessories Mobile heat extractor Thermal block Warranty and service Decontamination Certificate master All laboratory personnel designated to operate the instrument must carefully read the present instruction manual prior to starting work with the instrument.
This type of disinfection reduces the number of germs by at least The Leica CMUV is a powerful cryostat for routine as well as research applications in biology, medicine and industry. The instrument has been designed for rapid freezing and sectioning of tissue samples and is equipped with an automated disinfection system.
The instrument is not suitable for unattended storage of tissue material. The instrument may only be operated within the scope of its designated use as described above and as per the instructions given in this manual. All information given in this instruction manual applies only to the instrument type indicated on the title page. A name plate, indicating the instrument serial number, is attached to the back of the instrument. This instruction manual includes important instructions and information related to the operating safety and maintenance of the instrument.
The instruction manual is an important part of the product which must be read carefully before setting up and using the instrument and must always be kept with the instrument. The instrument incorporates the following safety features: safety handwheel and knife guards on the knife holders. To prevent adverse health effects from UV radiation, the UV disinfection cycle can be started only after the sliding window has been properly closed.
Closing the window activates the corresponding safety features. If necessary, move the handwheel slightly forth and back until the locking mechanism engages. To unlock, push the locking pin 1 to the left from the recess at the handwheel. Only rotate the handwheel when the refrigeration system is on and the cryochamber is cold. For every manipulation in the cryochamber, or upon changing a specimen when the knife or disposable blade is clamped, or during breaks, cover the cutting edge with the knife guard.
The knife holders CN and CS are equipped with knife guards; on the knife holder CE the glass plate of the anti-roll guide constitutes the knife guard please refer to the separate instruction manual for your knife holder. Prior to manipulating the knife and specimen, or changing the specimen or knife, and during breaks, always lock the handwheel! This instrument has been built and tested in accordance with the following safety regulations on electrical measuring, control, regulating and laboratory devices.
For current information about applicable standards, please refer to the CE declaration of conformity on our Internet site:. In order to maintain this condition and to ensure safe operation, the operator must observe the instructions and warnings contained in this instruction manual. The compressor requires a start-up current between 45 and 50 A. Therefore, the electric circuit at the place of installation must be inspected by an electrical engineer to ensure that it meets the requirements for a smooth operation of the instrument.
This waiting period is necessary to allow the compressor oil, which may have been displaced during transport, to return to its original position. Failure to comply with this can cause severe damage to the instrument. Therefore, do not touch it! The cutting edge is extremely sharp and can cause severe injury! Only shut the tap when draining the waste container! The instrument has been designed for UV disinfection.
Spray disinfection with Leica Cryofect is also possible, thanks to the special insulation of the microtome. Erst dann Desinfektion starten! Each new specimen is a potential source of contamination. Only use the cleaning agents and disinfectants specified in this instruction manual such as Leica Cryofect alcohol or common disinfectants based on alcohol!
Please contact Leica Biosystems for further details on adequate disinfection measures. Humidity inside will condense and freeze in the cold cryostat and thus cause malfunctions or damage. This can cause damage to the cooling system! Using fuses other than those specified by the manufacturer may cause severe damage to the instrument! If this happens, the lamp change must be completed by Technical Service. If any metallic mercury is released, handle it carefully and dispose of it properly.
If both disinfection indicator lights are blinking alternately, the UV tube must be replaced! Do not operate in rooms with explosion hazard! To ensure an adequate cooling capacity, the instrument must be set up with at least 10 cm distance from walls and furniture. High room temperatures and excessive air humidity affect the cooling capacity of the cryostat.
Please note the areas which are reinforced for transporting and grip the cabinet only at these locations Fig. The instrument must be transported in an upright position only. When the instrument is tilted, the compressor oil is displaced. Do not grip the cabinet at the lid.
Grip the cabinet only at the marked locations. The alignment of the adjustable feet is necessary to ensure an unobstructed drain of the quick freeze shelf defrosting water. Heat extractor, stationary Low temperature stabilizer for heat extractor, Parking station Optional clamping lever for object head Set of specimen discs Section waste tray Storage shelf, right Storage shelf, left Brush shelf Bottle of cryostat oil, type , 50ml Bottle of OCT-Compound, mounting medium for cryosectioning, ml To dismount, proceed in reverse order.
The handwheel including the fixing components are packed in the cardboard box for the accessories. The handwheel can be dismounted for transporting e. Please note that the compressor requires a start-up current between 45 and 50 A. Therefore, the electric circuit at the installation site must be inspected by an electrical engineer to ensure that it meets the requirements for a smooth operation of the instrument. A constant adequate power supply to the instrument must be ensured at all times.
Welcome to ManualMachine. We have sent a verification link to to complete your registration. Log In Sign Up. Forgot password? Enter your email address and check your inbox.